Wishing there was an easy way to monitor your blood glucose levels? With the new Accu-Chek Instant meter, it's as easy and fast as checking your phone messages. There is no setup required – just insert a test strip, apply a small blood sample and read your blood glucose result on the large display.
Reference: 11. ISO 15197:2013. In vitro diagnostic test systems- Requirements for blood-glucose monitoring system for self-testing in managing diabetes mellitus. 2. Accu-Chek Instant package insert. 3. Breitenbeck N. Brown, A: Assessment of a Blood Glucose Monitoring System for Self-Testing with Three Test Strips Lots following ISO 15197:2013/EN ISO 15197:2015. 4. Kocher S., et al., J Diabetes Sci Technol 2009;3(5):1136-1143. Disclaimer: Accu-Chek Softclix lancing device is for single patient use only. The same device must not be used for multiple patients.
FAD glucose dehydrogenase (GDH), electrochemical
4°C to 45°C
-25°C to 70°C
10 to 600 mg/dL
720 blood glucose results and 30 control tests results with time and date are also stores and can be viewed with external software Viewable on meter: last result and 7, 30 and 90 day test result averages.
Capillary, venous, arterial, and neonate
Capillary fill function
7, 30 and 90 days
Test strips remain stable up to expiry date printed on test strip vial, even after opening (test strip container must be closed tightly after each test strip is removed)
Two 3-volt lithium batteries (coin cell type CR2032 )
Micro USB cable (with computer)
90 seconds after performing a test, 15 seconds after a test strip is removed, or 5 seconds from the last test result screen.
Up to 3,094 m (0 to 10,150 ft.) above sea level
40g with battery
Get results in 3 steps: